The infrared temperature measuring device is the human body surface temperature screening instrument. The test results reflect the real-time temperature evaluation of objects or people in a specific environment. It is mobile, portable and light-weight, which can attach anywhere indoor or outdoor facilities.
Real time infrared thermo sensor to detect temperature, ready in 30 seconds after power on.
The distance sensor starts working when a person or object is in 20 cm to it. If the temperatureof person or object detected is ≥ 35.5 ℃, the voice "Normal temperature" is trigged; If it's lower than 35.5 ℃, or higher than the set value with alarm, the voice "Please check again" is trigged. When the ambient temperature is lower than 15 ℃,If thetemperatureof person or object detected is ≥ 34.5 ℃, the voice "Normal temperature" is trigged; If it's lower than 34.5 ℃, or higher than the set value with alarm, the voice "Please check again" is trigged.
Special attention: the distance sensor shall not be blocked, otherwise it will repeat the voice broadcast.
1. Power on and wait for 30 seconds, keep nothing in front of the temperature sensor.
2. Normal ambient temperature is 0 ° C - 30° C. When using it at - 10 ° C, you have to decrease the environment temperature compensation, 5° C per time, until the result difference between it and the forehead thermometer is less than ± 0.3° C. In case the results of many persons are low, it can also be calibrated by this method. When it is used in an environment more than 30° C , it is necessary to customize the sun protection and anti high temperature switch.
3. Measurement method: Measure forehead or wrist surface
4. Forehead temperature and wrist surface temperature can be measured without adjusting settings.
5. Measurement area: the center of forehead or the wrist covered by sleeve.
6. Standard measurement distance: 1cm-10cm. When measuring at a distance of 10-30cm, it is necessary
to adjust the environment temperature compensation, 5° C per time, until the result difference between it
and the forehead thermometer is less than ± 0.3° C.
7. Measurement time: 0.5 seconds. After the measurement, please leave quickly. Otherwise the
measurement will be repeated.
8. During the temperature measurement, influenced by the measurement speed, the temperature might
suddenly fluctuate up and down. In this case, please measure the body temperature 3-5 times
continuously, and take the average value as the final measurement result.
9. The instrument automatically detects the temperature of the object in 20cm in front of it. After measuring
the temperature, we have to move our hand or forehead away. If we keep close to the sensor for long time, it will stop working as a protection.4-4
Product weight: 13kg
Power consumption: less than 10W
Working temperature: -10 ~ 40 ℃
Operating humidity: 95%, non-condensing
External power supply: AC 100 ~ 240V, 50 / 60Hz
Temperature measurement speed: less than 1 second
Temperature measurement accuracy: ± 0.3,
Temperature measurement distance: 1-20CM
Product size: 1500(H)* 260(W)* 300(D) mm